Mechanical work equation for the heart
Work = pressure x stroke volume
Law of Laplace [sphere]
P = 2T / R or
T = (P x R) / 2
(pressure, tension, radius)
MAP of tall man standing (head, heart, feet)
Head: 52 mmHg
Heart: 90 mmHg
Feet: 203 mmHg
Relationship of MAP, CO, PVR
MAP = CO x PVR
Width of BP cuff compared to diameter of the arm
Width should be 20% larger than the diameter of the arm.
Modern BP cuffs are considered what & how do they work
-As BP decreases the needle oscillates (integrated electronics)
Drawbacks of automated oscillometers to measure BP (x2)
-2 min minimum interval
-Fails at BP < 50 mmHg
Identify the BP measuring device & explain how it works
-Uses LED plethysmography at fingertip & inflatable cuff to measure continuous BP at fingertip
Rate NS infuses in an arterial line
Frequency range for invasive BP monitor
0 to 40 Hz
Placing art line in brachial artery
a BAD idea
Resonance in art lines & how to prevent
Always occurs due to the inertia of the fluid column but as long as it is >40 Hz it does not interfere with the quality of the reading.
-Kept above 40 Hz by using stiff, short tubing.
Dampening in art lines & impact
Occurs with an air bubble or clot (they absorb the higher frequencies)
-usually negligible to MAP reading
PA catheter waveforms
Please allow access to your computer’s microphone to use Voice Recording.
We can’t access your microphone!
Click the icon above to update your browser permissions and try again
Reload the page to try again!
Press Cmd-0 to reset your zoom
Press Ctrl-0 to reset your zoom
It looks like your browser might be zoomed in or out. Your browser needs to be zoomed to a normal size to record audio.
Your microphone is muted
For help fixing this issue, see this FAQ.