Create an account
Comparison of Technical Factors Required
Single-Phase (x mAs)
Three-Phase, 6-Pulse (2/3x mAs)
Three-Phase, 12-Pulse (1/2x mAs)
When changing from single-phase to three-phase, six-pulse equipment,
two thirds of the original mAs are required to produce a radiograph with similar density
When changing from single-phase to 3-phase, 12-pulse equipment,
only one-half of the original mAs is required.
changing from single-phase to three-phase, 12-pulse equipment; therefore the new mAs should be
half the original
The amount of replenisher solution added as a film enters the automatic processor is related to the
1. size of the film.
2. number of films processed.
Replenishment is also essential for
maintaining each solution's level of concentration, to maintain solution activity and avoid chemical fog from exhausted solutions.
One way to determine the quantity of replenisher solution to be added is by
the length of the film entering the processor.
A microswitch initiates and terminates replenishment as
it senses the beginning and end of each film.
Greater latitude is available to the radiographer when using
1. high kVp factors.
2. a slow film-screen combination.
Which of the following adult radiographic examinations usually require(s) use of a grid?
Parts generally requiring the use of a grid include
the skull, spine, ribs, pelvis, shoulder, and femur.
osteomalacia, fibrosarcoma, and paralytic ileus (obstruction), result in a
decrease in body tissue density.
A radiograph exposed using a 12:1 ratio grid may exhibit a loss of density at its lateral edges because the
SID was too great
If the x-ray tube is angled significantly across the lead strips of a focused grid, there is
uniform loss of density (grid cutoff)
Insufficient or excessive distance with focused grids causes
loss of density (grid cutoff) along the periphery of the image.
An increase in exposure factors is usually required in which of the following circumstances?
The fact that x-ray intensity across the primary beam can vary as much as 45% describes the
anode heel effect
The anode heel effect becomes more pronounced as
the SID decreases, as IR size increases, and as target angle decreases.
Which of the following methods can be effectively used to decrease differential absorption, providing a longer scale of contrast in the diagnostic range?
1. Using high kVp and low mAs factors
2. Using compensating filtration
The photoelectric effect
the interaction between x-ray photons and matter that occurs at low kVp levels—levels that tend to produce short-scale contrast.
Which of the following function(s) to reduce the amount of scattered radiation reaching the IR?
1. Grid devices
2. Beam restrictors
Which portion of the characteristic curve would most likely represent a density of 1.0?
A characteristic curve is used to
predict the speed, contrast, and exposure latitude of a particular film emulsion.
A characteristic curve has three portions:
the toe, the straight-line portion (region of correct exposure), and the shoulder
follows the toe; portion of correct exposure and extends from about 0.25 to 2.5
Which of the following factors affect(s) both radiographic density and intensifying screen speed?
1. Thickness of phosphor layer
2. Type of phosphors used
Rare earth-type phosphors
absorb x-rays more efficiently and convert their energy into fluorescent light; they therefore affect both screen speed and radiographic density
The thickness of the phosphor layer affects
speed and density similarly: As the thickness of the phosphor layer increases, speed and density increase.
The spongy layer behind each intensifying screen helps ensure
good screen-film contact and therefore good recorded detail. The spongy layer is unrelated to radiographic density.
Which of the following can affect histogram appearance?
1. Centering accuracy
2. Positioning accuracy
3. Processing algorithm accuracy
The CR laser scanner recognizes the various tissue density values and constructs a
representative gray-scale histogram
Histogram analysis and use of the appropriate LUT together function to
produce predictable image quality in CR.
Degree of accuracy in positioning and centering
can have a significant effect on histogram appearance (as well as patient dose)
Change is affected in
average exposure level and exposure's latitude; these changes will be reflected in the images informational numbers ("S number," "Exposure Index," etc)
Other factors affecting histogram appearance and, therefore, these informational numbers, include
selection of the correct processing algorithm (e.g., chest vs femur vs cervical spine), changes in scatter, SID, OID, and collimation.
If a 6-inch air gap (OID) is introduced between the part and IR
much of the scattered radiation emitted from the body will not reach the IR
A technique chart should include which of the following information?
1. Recommended SID
2. Grid ratio
3. Screen-film combination
Technique charts are
exposure factor guides that help technologists produce radiographs with consistent density and contrast
They suggest a group of exposure factors to be used at
a particular SID with a particular grid ratio, screen-film combination, focal spot size, and central ray angulation.
In which of the following ways can higher radiographic contrast be obtained in abdominal radiography?
1. By using lower kilovoltage
2. By using a contrast medium
3. By limiting the field size
Higher contrast is
shorter scale contrast; it is present in an image that has few shades of gray between white and black.
High radiographic contrast also results when
radiographing anatomic parts that have high subject contrast, such as the chest.
The abdomen has low
subject contrast, and therefore abdominal radiographs will tend to have very low contrast unless technical factors are selected to increase contrast.
To better demonstrate high contrast within a viscus
a contrast medium such as barium, iodine, or air can be used.
Restricting the size of the field will also function to
increase contrast because less scattered radiation will be generated.
Which of the following pathologic conditions are considered additive conditions with respect to selection of exposure factors?
A grid is usually employed
1. when radiographing a large or dense body part.
2. when using high kilovoltage.
The latent image is
the invisible image produced within the film emulsion as a result of exposure to radiation
How can the radiographer reduce the amount of scattered radiation generated during a radiographic examination?
1. Use optimum kVp.
2. Collimate closely.
The amount of scattered radiation generated in a given exposure increases as three factors increase:
the kVp, the field size, and the thickness and density of the tissues.
Geometric unsharpness is influenced by which of the following?
1. Distance from object to image
2. Distance from focus to object
3. Distance from focus to image
If single-emulsion film was loaded into its cassette with the emulsion facing away from the intensifying screen, the resulting image would demonstrate
Single-emulsion film is used
with a cassette that has a single intensifying screen and provides better detail than typical double-emulsion film and two-screen cassettes
Single-emulsion film has an antihalation backing that
efficiently absorbs reflected (crossover) light.
if the film were loaded with the antihalation side against the light-emitting screen, the film emulsion would
receive very little exposure
The processor rollers that are out of solution and function to transfer the film from one solution to another are the
are located at the bottom of each solution tank and function to direct the film from a downward to an upward motion
also called deflector plates, serve to keep the film on its proper course by directing or guiding it around corners.
The crossover rollers
are located at the top of the processor, out of the solution, and direct the film from one solution tank to the next. These are the racks that need daily cleaning to avoid chemical or emulsion buildup on their surface. Chemical or emulsion buildup on roller surfaces can cause film artifacts.
If the x-ray beam is not centered to the grid, or if the x-ray tube and grid surface are not parallel (level), there will be a fairly uniform decrease in
radiographic density across the entire image.
if the grid is not used within its recommended SID (focus) range (ie, if the SID is too great or too little), there will be
a decrease in density at the periphery of the image.
is said to be the greatest enemy of recorded detail because it completely obliterates image sharpness
Poor screen-film contact
reduces recorded detail because of the degree of light diffusion in the areas of poor contact. Areas of poor contact appear blurry.
Which of the following conditions require(s) a decrease in technical factors?
If the quantity of black metallic silver on a particular radiograph is such that it allows 1% of the illuminator light to pass through the image, that image has a density of
OID is used to effect
an increase in contrast in the absence of a grid, usually in chest radiography
The x-ray tube anode is designed according to the
line focus principle, that is, with the focal track beveled
This allows a larger actual focal spot to project a
smaller effective focal spot, resulting in improved recorded detail with less blur
because of the target angle,
penumbral blur varies along the longitudinal tube axis, being greater at the cathode end of the image and less at the anode end of the image.
Radiographic contrast is described as
the difference between densities, or scale of grays, in the radiographic image.
the function of grids is to
collect scattered radiation, they serve to shorten the scale of contrast.
Beam restrictors function to
limit the x-ray field size, thereby reducing the production of scattered radiation and shortening the scale of contrast.
Focal spot size
one of the geometric factors affecting recorded detail; it has no effect on the scale of contrast.
When the collimated field must extend past the edge of the body, allowing primary radiation to strike the tabletop, as in a lateral lumbar spine, what may be done to prevent excessive radiographic density due to undercutting?
Use lead rubber to absorb tabletop primary radiation
When the primary beam is restricted to an area near the periphery of the body, sometimes
part of the illuminated area overhangs the edge of the body
If the exposure is then made, scattered radiation from the tabletop (where there is no absorber) will undercut the part, causing
excessive image density.
If, however, a lead rubber mat is placed on the overhanging illuminated area, most of this scatter will be
Improper spectral matching between rare earth intensifying screens and film emulsion results in
If, for example, a blue-sensitive emulsion were matched with green-emitting screens, the resulting radiograph would be
underexposed because the blue-sensitive film emulsion was not responsive to the green-emitting phosphors.
In which of the following examinations would a cassette front with very low absorption properties be especially desirable?
Because mammographic techniques operate at very low kVp levels, the cassette front material
becomes especially important.
The use of soft, low-energy x-ray photons is the underlying principle of
mammography; any attenuation of the beam would be most undesirable
Special plastics that resist impact and heat softening, such as polystyrene and polycarbonate, are frequently used as
cassette front material
The presence of dust or scratches on intensifying screens will cause
decreased density in those areas of the image
If intensifying-screen phosphors are covered with dust
either they will not fluoresce or their fluorescence will not reach the IR emulsion
if the screen is scratched and phosphors are removed,
there will be no fluorescence to expose the IR.
The mAs selected is directly proportional to
radiographic density (ie, if the mAs is cut in half, radiographic density will be halved)
Although SID affects exposure rate (according to the inverse square law of radiation), and therefore affects density (according to the density maintenance formula), it is not used to regulate
A grid is
a thin wafer placed between the patient and the IR to collect scattered radiation. It is made of alternating strips of lead and a radiolucent material such as plastic or aluminum.
As the kilovoltage is increased
more penetration will occur and a greater range of densities (grays) will be apparent in the image
In addition, as the kVp and scale of grays increase,
the exposure latitude increases; the "margin for error" in technical factors becomes greater.
According to the line focus principle, an anode with a small angle provides
1. improved recorded detail.
2. improved heat capacity.
The line focus principle illustrates that as the target angle decreases,
the effective focal spot decreases (providing improved recorded detail), but the actual area of electron interaction remains much larger (allowing for greater heat capacity).
It must be remembered, however, that a steep (small) target angle increases
the heel effect, and part coverage may be compromised
The effect that differential absorption has on radiographic contrast of a high subject contrast part can be minimized by
1. using a compensating filter.
2. using high-kVp exposure factors.
Differential absorption refers to
the different attenuation, or absorption, properties of adjacent body tissues
Two parts with widely differing absorption characteristics will produce
a high radiographic contrast.
exposure factors that would properly expose one part will
severely overexpose or underexpose the neighboring part (as with lungs vs thoracic spine)
This effect can be minimized by the use of
a compensating filter or by the use of high kilovoltage (for more uniform penetration).
Increased collimation is important in
the control of patient dose and scattered radiation, not differential absorption.
Which of the following radiographic accessories functions to produce uniform density on a radiograph?
can be wedge-shaped (with the thicker part of the wedge paralleling the thinner body part), thus compensating for greater or lesser tissue densities (as in a large decubitus abdomen)
An overall image density arising from factors other than the light or radiation used to expose the image is called
Anything other than intensifying screen light or primary x-radiation is
undesirable in terms of image exposure
Log relative exposure
the amount of exposure required to produce a given density as measured on the sensitometric graph
anything foreign to the image; the term could include fog, but it also covers many physical interferences.
As kVp is increased,
x-ray photons begin to interact with atoms of tissue via the Compton scattered interaction
While Compton scatter reduces patient dose, compared to photoelectric interactions
it can pose a significant radiation hazard to personnel during fluoroscopic procedures.
Which of the following is most likely to occur as a result of using a 30-inch SID with a 14 x 17-inch IR to radiograph a fairly homogeneous structure?
Density variation between opposite ends of the IR
Because shorter SIDs and larger IR sizes require greater divergence of the x-ray beam to provide coverage,
the anode heel effect will be accentuated.
he type of chemistry used in the automatic processor and especially the temperature of the solution can have a big impact on
the resulting image contrast.
Which of the following pathologic conditions require(s) a decrease in exposure factors?
3. Multiple myeloma
a chronic pulmonary disease characterized by an increase in the size of the air-containing terminal bronchioles.
Please allow access to your computer’s microphone to use Voice Recording.
Having trouble? Click here for help.
We can’t access your microphone!
Click the icon above to update your browser permissions and try again
Reload the page to try again!Reload
Press Cmd-0 to reset your zoom
Press Ctrl-0 to reset your zoom
It looks like your browser might be zoomed in or out. Your browser needs to be zoomed to a normal size to record audio.
Please upgrade Flash or install Chrome
to use Voice Recording.
For more help, see our troubleshooting page.
Your microphone is muted
For help fixing this issue, see this FAQ.
Star this term
You can study starred terms together