Create an account
Which of the following is (are) characteristics of a 16:1 grid?
1. It absorbs more primary radiation than an 8:1 grid.
2. It has less centering latitude than an 8:1 grid.
3. It is used with higher kVp exposures than an 8:1 grid.
In which type of equipment does kVp decrease during the actual length of the exposure?
1. Battery operated mobile equipment
2. Fixed x-ray equipment
Mobile x-ray machines
are compact and cordless and are either the battery-operated type or the condenser discharge type
Condenser discharge mobile x-ray units
do not use batteries; this type mobile unit requires that it be charged before each exposure
the stored energy is used
to operate the x-ray tube only. Because this machine does not carry many batteries, it is much lighter and does not need a motor to drive or brake it
The major disadvantage of the capacitor discharge unit is
that, as the capacitor discharges its electrical charge, the kVp gradually decreases throughout the length of the exposure, therefore limiting tube output and requiring recharging between exposures.
A light-absorbing dye is frequently incorporated during the manufacture of screens to
reduce the diffusion of fluorescent light
Remnant radiation emerging from the patient causes
fluorescence of the cassette's intensifying screens.
. Fluorescent light that is not perpendicular produces
a blur, resulting in geometric unsharpness and decreased resolution.
During manufacture, special dyes can be added to the active layer that will
absorb much of the diffused fluorescence and allow a greater percentage of the image density to be created by more perpendicular fluorescent light.
Although the stated focal spot size is measured directly under the actual focal spot, focal spot size in fact varies along the length of the x-ray beam. At which portion of the x-ray beam is the effective focal spot the largest?
At the cathode end
X-ray tube targets are constructed according to the
line focus principle—the focal spot is angled (usually 12° to 17°) to the vertical
As the actual focal spot is projected downward,
it is foreshortened; thus, the effective focal spot is smaller than the actual focal spot
As the focal spot is projected toward the cathode end of the x-ray beam,
it becomes larger and approaches its actual size.
As the exposed film enters the processor from the feed tray, it first enters
the developer section, where the emulsion's exposed silver bromide crystals are reduced to black metallic silver.
The film then enters the
fixer, where the unexposed silver grains are removed from the film by the clearing agent (hypo).
The film then enters the
wash section, where chemicals are removed from the film to preserve the image.
What is the name of the device that functions to expose a film with an optical step wedge having a number of densities ranging from white to black?
Two devices that are required for quality assurance evaluation purposes are
sensitometer and a densitometer
is a device that functions to expose a film; when developed, that film will illustrate an optical step wedge with a number of optical densities ranging from white through several shades of gray to black.
or (usually aluminum) step wedge, can be used to demonstrate the effect of kVp on radiographic contrast.
Excessive emulsion swelling and chemical fog can be caused by
increased solution temperature or insufficient replenishment.
What determines the amount of fluorescent light emitted from a fluorescent screen?
1. Thickness of the active layer
2. Type of phosphor used
3. kV range used for exposure
As the kVp level is increased
so is the amount of fluoroscopic light emitted by intensifying screen phosphors
When green-sensitive rare earth screens are properly matched with the correct film, what type of safelight should be used in the darkroom?
It is essential that the darkroom safelight filter color is correctly matched with
the type/sensitivity of film emulsion being used.
This circuit provides current and voltage to
heat the filament to incandescence, at which time it undergoes thermionic emission (the liberation of valence electrons from filament atoms).
The greater the number of electrons flowing between the cathode and the anode
the greater the tube current (mA)
Rectification (single- or three-phase)
is the process of changing alternating current to unidirectional current.
indicates optical density by providing a digital readout of the quantity of light transmitted through an x-ray image
is a device used to produce a consistent gray scale exposure on film, usually in conjunction with processor sensitometry
H&D (Hurter & Driffield) curves
are sensitometric curves of a film emulsion's response to light and radiation.
A misaligned guide shoe in the turnaround assembly will create
evenly spaced minus-density lines as a result of emulsion scratching
appear as minus-density lines resulting from improper positioning or centering of the grid or x-ray tube
starts with a clear polyester base that serves as support for the emulsion. Applied next is an adhesive layer that functions to hold the emulsion to the base. Next is the emulsion, consisting of silver halide grains suspended in gelatin. Finally, a supercoat of clear hard gelatin is applied as an antiabrasive layer.
Please allow access to your computer’s microphone to use Voice Recording.
Having trouble? Click here for help.
We can’t access your microphone!
Click the icon above to update your browser permissions and try again
Reload the page to try again!Reload
Press Cmd-0 to reset your zoom
Press Ctrl-0 to reset your zoom
It looks like your browser might be zoomed in or out. Your browser needs to be zoomed to a normal size to record audio.
Please upgrade Flash or install Chrome
to use Voice Recording.
For more help, see our troubleshooting page.
Your microphone is muted
For help fixing this issue, see this FAQ.
Star this term
You can study starred terms together