Create an account
Describe aperture diaphragm?
Flat sheet of metal, usually lead, with a hole cut in center and attached to xray tube port
The size of the opening for an Aperture diaphragm depends on?
Desired field size, SID, distance from the aperture diaphragm to the focal spot
Disadvantage of aperture diaphragm?
Increase in penumbra- geometric unsharpness around the periphery of the image
- due to diaphragms close proximity to the tube port
Describe a cone?
Has an extension that flares or diverges, with upper diameter smaller then the bottom flared end
Does not flare, has same diameter at the bottom of the extension as it has on the top. Possible extension sleeve which can expand or collapse to vary the degree of beam restriction
Advantages of cones and cylinder?
Inexpensive and simple
-Cylinder reduces penumbra and off-focus radiation
Advantages of Collimator?
Provides a light source to outline exposure field and provides crosshair to identify the center of the beam
Collimator consist of?
Lead shutters at right angles to one another, which adjust to infinite number, attached to tube head
The purpose of a collimators shutters?
Regulate field size.
-Bottom shutter reduce penumbra along periphery, has 2 pairs of shutters for field length & width
-Upper shutters (closes to tube) help reduce amount of off-focus (stem) radiation reaching IR by absorbing radiation
Design of collimator (light field)?
Light field provided by mounting mirror in path of xray beam at angle of 45 degrees. Light source and xray source must be equidistant from each other
What do PBL devices do?
When cassette is loaded, sensing device collimates lead shutter down to size of loaded cassette
Use of collimator results in some xray beam filtration due to?
Primary beam passing through the mirror in the collimator.
added filtration= 1 mm Al
For the light field in the xray field to correspond accurately
Xray tube focal spot and light bulb must be exactly at the same distance from the center of the mirror
National Council on Radiation Protection (NRCP) guidelines state the collimators must be accurate within?
2% fo the SID
For PBL, the NCRP requires that the guidelines be accurate to within
3% of the SID for a singe side, and within 4% of the SID for all total sides
Please allow access to your computer’s microphone to use Voice Recording.
Having trouble? Click here for help.
We can’t access your microphone!
Click the icon above to update your browser permissions and try again
Reload the page to try again!Reload
Press Cmd-0 to reset your zoom
Press Ctrl-0 to reset your zoom
It looks like your browser might be zoomed in or out. Your browser needs to be zoomed to a normal size to record audio.
Please upgrade Flash or install Chrome
to use Voice Recording.
For more help, see our troubleshooting page.
Your microphone is muted
For help fixing this issue, see this FAQ.
Star this term
You can study starred terms together