Create an account
anatomically programmed radiography (APR)
Computerized controls that are programmable so that exposure factors from a conventional technique chart can be made available from within the computer.
automatic exposure control (AEC)
An electronic circuit within the x-ray machine that automatically terminates the exposure time when a predetermined quantity of x-rays has been detected.
A manually set time that insures that an exposure will be terminated should the automatic exposure control fail.
More sophisticated devices designed for use with three-phase and high-frequency generators; more capable of ultrashort exposure times; time settings available will be expressed in decimals; most accurate.
Closes the circuit, allowing current to flow through the primary side of the step-up transformer.
A device that terminates the exposure by opening a switch after a preset time has elapsed.
In this form of rectification, four diodes are employed in the circuit so the current can be "redirected" during the negative half of the electric cycle so that the current will flow in the same direction during both halves of the cycle; allows x-rays to be formed during the whole cycle of current.
In this form of rectification, diodes (one or two) are added to prevent the damage that might occur if the tube fails to provide adequate self-rectification.
high-frequency x-ray generator
Employs special "inverter circuits"; provides a less expensive means of increasing the efficiency of x-ray production; can produce a near constant beam of x-rays.
An enclosed volume of dry air connected in a circuit; located between the patient and the IR.
Measures the voltage output from the autotransformer; predicts the voltage across the x-ray tube.
line voltage compensator
Ensures that the outgoing voltage is accurate, regardless of fluctuation in the incoming line; located on the primary (input) side of the autotransformer.
All of the devices of the ____ are located at the control console, with exception of the step-up transformer.
milliammeter (mA meter)
Measures the current flowing in the high-voltage circuit during an exposure; connected at the midpoint of the secondary side of the step-up transformer.
Makes the current "right" for x-ray production by ensuring that it flows only in the right direction, in this case from filament to target.
A unit used to change the AC into direct current (DC) so that it flows in only one direction.
Necessary to make an exposure; also called "Prep", "Ready," or "Standby"; two functions: when activated it starts the rotating anode which creates the electrons and when held in the "on" position for a few seconds, it will indicate that the tube is ready for an exposure.
The x-ray tube itself controls the direction of current flow; current flows only during the positive portion of the electric cycle, when the anode is positive and the cathode is negative.
Produce a pulsating current that alternates from positive to negative during each electric cycle resulting in a pulsating x-ray beam.
Controlled by a small electric motor rotating at 60 revolutions/sec; time settings available will be expressed in fractions in multiples of 1/60 sec.
Alternating current is generated in three overlapping cycles that produce the waveform; generates x-rays more efficiently than single-phase current.
The subcircuit between the alternating current (AC) power supply (1) and the primary (input) side of the high-voltage (step-up) transformer (11).
The unit where the operator sets all the exposure techniques, such as kVp, mA, and T/s.
main switch, line meter, line voltage compensator, autotransformer, kVp selector, kVp meter, contacts for autotransformer, exposure switch, and exposure time
Located in or on the control console.
Small variations in the incoming ____ may cause large variations in the kVp to the x-ray tube.
functions of the autotransformer
Provides the means for the kVp selection, it provides compensation for fluctuations in the incoming line voltage, and it supplies power to other parts of the x-ray circuit.
autotransformer's primary purpose
To vary the voltage to the primary side of the step-up transformer.
When the exposure switch closes the circuit, the current is induced to flow through the ____ of the transformer, creating voltage across the x-ray tube.
The subcircuit of the x-ray circuit that is divided into two parts by the step-down transformer.
purpose of the filament circuit
To supply and control the heat required by the x-ray tube filament for thermionic emission of electrons.
The ____ of the filament circuit begins and ends with the contacts on the autotransformer.
Current flows from autotransformer to mA selector to primary side of step-down transformer and back to the autotransformer. Secondary side of step-down transformer conducts current through x-ray tube filament and back to the secondary side.
The step-down transformer reduces the ____ on the secondary side, providing an appropriate current to heat the filament.
This circuit begins and ends with the secondary side of the step-up transformer.
autotransformer, step-down transformer, mA selector, x-ray tube filament, and focal spot selector
Located in the filament circuit.
secondary side of the step-up transformer, x-ray tube, the milliammeter, and the rectifier unit
Located in the high-voltage circuit.
A set multiple by which a transformer always increases or decreases the incoming voltage.
Provides safety from high-voltage shock when the meter is mounted in the control panel.
With self-rectification, the ____ of the x-ray tube prevents current from flowing from anode to cathode during the negative phase of the electric cycle.
Prevent "backflow" of current during the negative half of the electric cycle, and it is eliminated.
placed between the anode and the transformer
Placement if one diode is used in a half-wave rectification.
placed between the cathode and the transformer
Placement of second diode if two are used in a half-wave rectification.
self rectification, half-wave rectification, and full-wave rectification
Produced by single-phase electric power.
When this current is rectified, its waveform has the appearance of a "ripple" with no real low points.
40% to 50%
In a three-phase current, exposure times are decreased ____, because more x-rays are produced for the same kVp and results in lower patient dose.
Produces a near-constant voltage waveform results in even less exposure time and lower patient dose compared to three-phase.
"conventional" x-ray control panel
Operated by means of knobs and switches. Dials or meters indicate the settings.
computerized x-ray control panel
Have buttonlike controls and digital readouts; automatically perform some functions.
kVp, mA, exposure time, and filament/focal spot size and indicate the current settings
All control panels provide some means for selecting ____.
Controls the power to the control panel and the entire x-ray generator; corresponds to the main power switch in circuit diagram.
Older x-ray machine control panels include a meter to monitor fluctuations in ____ and a control knob or button to adjust for ____ compensation; modern control units monitor ____ and compensate for fluctuations automatically.
Provide variable mA that is computer controlled according to the selection of the desired mAs and focal spot size.
reproducibility of exposure, mAs reciprocity, and mA and exposure time linearity
Quality control check tests regularly performed on the mA settings to ensure accuracy.
APR automatic mode
The radiographer enters the examination and the body part measurement, and the computer selects the correct exposure factors.
depends on back-up timer (may be synchronous or electronic)
Automated timer's minimum exposure time.
The exposure is terminated when a specific quantity of current has been measured in this circuit.
Both phototimers and ionization chambers usually have ____ detectors so that limited operators can select the location or locations within the radiation field where the radiation quantity will be measured.
The exposure timer is set to a time that is ____ than the anticipated exposure time because if the back-up time setting is not long enough, the back-up timer will terminate the exposure prematurely.
X-ray operators should voluntarily set the backup time at ____ more than the anticipated exposure.
Premature release of either the rotor switch or the exposure switch will abort the ____ before it is complete.
check line voltage; adjust if needed
First step in setting control panel and making an exposure; traditional x-ray control.
select milliamperes (mA)
Next step (after checking line voltage) in setting control panel and making an exposure; traditional x-ray control.
select exposure time
Next step (after selecting mA) in setting control panel and making an exposure; traditional x-ray control.
select kVp major first, then minor
Next step (after selecting exposure time) in setting control panel and making an exposure; traditional x-ray control.
set Bucky switch
Next step (after selecting kVp) in setting control panel and making an exposure; traditional x-ray control.
activate and hold rotor switch
Next step (after setting Bucky) in setting control panel and making an exposure; traditional x-ray control.
on signal, activate and hold exposure switch
Next step (after activating rotor switch) in setting control panel and making an exposure; traditional x-ray control.
observe exposure indicator to validate exposure and to determine when it is complete
Next step (after activating exposure switch) in setting control panel and making an exposure; traditional x-ray control.
release rotor and exposure switches
Last step in setting control panel and making an exposure (traditional x-ray control).
primary cause of problems that may shorten tube life
The anode of the tube accumulates enormous heat during exposures.
A ____ should be warmed up before any large exposure; an excessive exposure on a ____ may cause the anode to crack and to fail.
three exposures - 30 secs apart at a setting of 200 mA, 0.5 sec, and 80 kVp
Warm-up procedure in the absence of an established one.
A rapid series of large exposures or a single excessive exposure may damage the tube by melting the ____ of the focal track.
If the point of the tube rating chart is below the curved line that represents the mA setting, the exposure is ____.
If the point of the tube rating chart is above the curved line that represents the mA setting, the exposure is ____.
kVp x mA x time
Formula for how heat units (HU) produced by an exposure is determined; used for single phase generators.
Do not activate the ____ before you are completely ready to make an exposure; increases wear on both bearings and filament, shortening tube life.
Please allow access to your computer’s microphone to use Voice Recording.
Having trouble? Click here for help.
We can’t access your microphone!
Click the icon above to update your browser permissions and try again
Reload the page to try again!Reload
Press Cmd-0 to reset your zoom
Press Ctrl-0 to reset your zoom
Please upgrade Flash or install Chrome
to use Voice Recording.
For more help, see our troubleshooting page.
Your microphone is muted
For help fixing this issue, see this FAQ.
Star this term
You can study starred terms together